Serinus ® Cal 3000 OZONE TRANSFER STANDARD Reliability, accuracy and precision are hallmarks of ACOEM Ecotech’s product range. It’s why Ecotech’s Calibrators have long been trusted worldwide. The Serinus ® Cal 3000 includes an ozone photometer for the accurate creation and delivery of ozone concentrations when calibrating ozone analysers. The photometer accurately and continuously measures the ozone concentration to control output of the internal ozone generator, providing an accurate ozone source for routine ozone calibration. FEATURES • New Serinus ® Cal user interface makes all functions simple and intuitive • In-built photometer and ozone generator for use as a Level 3 ozone transfer standard • Ozone photometer based on the reliable and proven technology used in the Serinus ® 10 gas analyser. DILUTION & SPAN FLOWS Dilution Gas Inputs: 1 standard 100 - 200 kPa (g) (2 optional) Source Gas Inputs: 4 standard 100 - 200 kPa (g) (8 optional) Dilution Mass Flow Controller: 10 SLPM, 0 Deg, 1 ATM (std), 1 SLPM, 2 SLPM, 5 SLPM or 20 SLPM (optional), 2nd MFC (optional) Source Mass Flow Controller: 50 SCCM, 0 Deg, 1 ATM (std), 10 SCCM, 20 SCCM, 100 SCCM, 500 SCCM or 1 SLPM, 2 SLPM (optional), 2nd MFC (optional) Flow Accuracy (Constant Temp): Within 1 % of full scale Flow Repeatability: Within 0.15 % of full scale Linearity: Within 0.15 % of full scale Operating Gas Pressure: 100 - 200 kPa Zero Drift: < 0.6 % per year Response Time: < 5 seconds Output Manifold: 4 output ports standard Dilution Ratio: Variable 10:1 to 2000:1 (std configuration) Case Dimensions Rack length: 597 mm (23.5”) (front to rear) Total length (with latch release): 638 mm (25.1”) Chassis width: 418 mm (16.5”) Front panel width: 429 mm (16.9”) Chassis height: 163 mm / uses 4RU (6.4”) Front panel height: 175 mm (6.9”) Weight: 23.8 kg Specifications subject to change without notice. Ecotech & Air 1000 are trademarks or registered trademarks of ECOTECH Pty Ltd in the United States and/or other countries. © 2018 ECOTECH Pty Ltd. All rights reserved. 20180823
"Very efficient. Competitive pricing. The reminders, I’m so busy in my job that it’s so great to have a reminder to do something. It really helps me.
I’m very happy".
- Tim Cannon – Research and Development Manager; Pecan Engineering Pty Ltd