Serinus ® Cal 1000 OZONE TRANSFER STANDARD The Serinus ® Cal 1000 performs gas dilution calibrations and is used in conjunction with regulatory traceable gases and a zero air generator. The most cost effective solution, the Serinus ® Cal 1000 allows the most up-to-date calibrations technology to be available to all. DILUTION & SPAN FLOWS Dilution Gas Inputs: 1 standard 100 - 200 kPa (g) (2 optional) Source Gas Inputs: 4 standard 100 - 200 kPa (g) (8 optional) Dilution Mass Flow Controller: 10 SLPM, 0 Deg, 1 ATM (std), 1 SLPM, 2 SLPM, 5 SLPM or 20 SLPM (optional), 2nd MFC (optional) Source Mass Flow Controller: 50 SCCM, 0 Deg, 1 ATM (std), 10 SCCM, 20 SCCM, 100 SCCM, 500 SCCM or 1 SLPM, 2 SLPM (optional), 2nd MFC (optional) Flow Accuracy (Constant Temp): Within 1 % of full scale Flow Repeatability: Within 0.15 % of full scale Linearity: Within 0.15 % of full scale Operating Gas Pressure: 100 - 200 kPa Zero Drift: < 0.6 % per year Response Time: < 5 seconds Output Manifold: 4 output ports standard Dilution Ratio: Variable 10:1 to 2000:1 (std configuration) FEATURES • Performs single and multi-point calibrations using the Serinus ® Calibrator’s precision mass flow controllers, for utmost confidence in the results • Performs calibration manually through user interface or remotely via RS232, USB networks or digital inputs • Designed and developed with wide-ranging customer input for easy, intuitive, accurate use. Case Dimensions Rack length: Total length (with latch release): Chassis width: Front panel width: Chassis height: Front panel height: Weight: 597 mm (23.5”) (front to rear) 638 mm (25.1”) 418 mm (16.5”) 429 mm (16.9”) 163 mm / uses 4RU (6.4”) 175 mm (6.9”) 18.2 kg Specifications subject to change without notice. Ecotech & Air 1000 are trademarks or registered trademarks of ECOTECH Pty Ltd in the United States and/or other countries. © 2018 ECOTECH Pty Ltd. All rights reserved. 20180823
Very efficient. Competitive pricing. The reminders, I’m so busy in my job that it’s so great to have a reminder to do something. It really helps me.
I’m very happy.
- Tim Cannon – Research and Development Manager; Pecan Engineering Pty Ltd