Serinus ® Cal 300 OZONE TRANSFER STANDARD The Serinus ® Cal 300 includes an in-built ozone generator and internal zero-air source, as well as a compact reference photometer, providing precise feedback control for accurate ozone generation and measurement in the one unit. Ozone generation and measurement are automatically corrected for gas temperature and pressure changes,and can be displayed in units of ppm, ppb, ppt, ug/m3, mg/m3, ng/m3. FEATURES • Self contained transfer standard includes ozone photometer, ozone generator and zero air source within the one unit • Capable of measuring and generating accurate ozone calibration levels (Photometer accuracy: 1.0 ppb or 1 % of reading, ozone generator output 40 to 1000 ppb at 4 LPM) • Up to 16 sequences including 32 discreet points • Allows users to perform inter-station calibrations without disrupting station plumbing • Includes its own zero air generator and inbuilt ozone generator • Capable of measuring internally generated ozone or externally generated ozone enabling station ozone generators to be easily calibrated • Generation of ozone concentrations through internally stored calibration curve when feedback is disabled • Ensures repeatable and accurate readings by automatically compensating for temperature, pressure and flow variations • Integral output gas manifold, allows up to 4 instruments to be calibrated simultaneously • USB memory stick for data storage which can be retrieved via RS232, USB interface or the optional Ethernet connection • Programming, viewing, downloading and emailing of data is made even simpler using Bluetooth connectivity, combined with the ‘Ecotech Serinus ® Remote’ Android app. SPECIFICATIONS Range: • 0 - 3000 ppb ozone. User interface: • Full numeric keypad on front panel or remotely via computer • Large graphic LCD display, with unit selection, ppm, ppb, ppt, ug/ m3, mg/m3, ng/m3. Noise: • < 0.25 ppb or 0.1 % of reading RMS whichever is greater, with Kalman filter active. Lower detectable limit: • 0.5 ppb with Kalman filter active. Precision: •
"Very efficient. Competitive pricing. The reminders, I’m so busy in my job that it’s so great to have a reminder to do something. It really helps me.
I’m very happy".
- Tim Cannon – Research and Development Manager; Pecan Engineering Pty Ltd